Patient's Reference Guide
This guide provides the essential information you should know about Automator 2000 in a question- answer format.
What safeguards prevent Automator 2000 from distracting faster than it is supposed to?
Automator 2000 was engineered to provide highly precise adjustments. Through the window in the top of the device you will see a series of holes around the perimeter of the gear which directly controls the position and adjustment of the telescoping rod. A sensor marks the progress of these holes as the gear turns. If the progress of the gear does not maintain accuracy within 1/72nd mm than an alarm is sounded.
Further software and circuitry safeguards have been engineered into the device to prevent the motor from running when it is not supposed to and to confirm that the motor is operating under normal parameters.
Are there any maintenance requirements for Automator 2000?
Once Automator 2000 is initiated the device should require little maintenance. It is important, however, to periodically confirm the following:
How do I know when the battery power will run out?
The batteries in Automator 2000 are designed to last more than four months (when the rate is set to 1 mm per day). Most users will never see the device go to low power, however, should the batteries run low, an alarm will sound. Instead of the green “run” light flashing every 15 seconds, both the run light and red “alarm” lights will flash. The device will continue to run for only a few days after the low power alarm, so the patient should notify Autogenesis and their doctor immediately to get a replacement device if necessary.
How do I know if the device(s) is/are running?
In addition to noticing the extension of the telescoping rod, progress may also be seen through the window on the top of each device. Notice that the gear under the window has markings just like the nut on the telescoping rod that is used for manual turns. (1 dot, 2 dots, 3 dots, and 4 dots- each spaced 90º apart) When set for forward motion the gear will turn across the period of a day to reveal the dots increasing from 1 to 4 dots.
The red and green lights on the front of the device also confirm the operation of Automator 2000. During normal operation in run mode the green run light will flash every 15 seconds. The patient will also periodically hear the brief hum of the motor in each Automator. The following chart describes how frequently the device makes 1/360mm adjustments:
Rate Setting Adjustment Interval Rate Setting AdjustmentInterval
0.5 mm/day 8 Minutes 1.25 mm/day 3.5 Minutes
0.75 mm/day 6 Minutes 1.5 mm/day 3 Minutes
1.0 mm/day 4 Minutes 2.0 mm/day 2 Minutes
4.0 mm/day 1 Minute
Finally, the user may run a self diagnostic at any time by pressing the control button. If the run light flashes, the device is operating as programmed. If no lights flash and no chirp is heard than the device is in sleep mode.
If an Automator 2000 is alarming, what should I do?
Automator 2000 is designed to maintain accuracy within 1/72 mm. If the device is unable to maintain this high level of accuracy than an alarm will sound. It is likely, therefore, during some portion of the procedure that an alarm will be heard. If a device alarms press the control button for three seconds to silence the alarm. Continue to hold the button. An acknowlegement chirp shall be heard, however the button must be held down for an additional 8-10 seconds until the green LED starts flashing. Once the green LED is flashing release the button. Now press the button for 1 second. The device will resume operation. After a moment additional beeps will be hear representing the rate at which the device is set. Now the device is running again.
What should I do if Automator 2000 alarms more than once, or I am unable to reset the device?
If Automator 2000 alarms repeatedly or there is difficulty restarting the device, you should call Autogenesis, Inc. (See phone numbers below.) A representative at Autogenesis will assist you with resetting the device and determining if there is a problem that requires attention.
Can I go swimming or take a shower with Automator 2000 on?
Automator 2000 is designed to be waterproof to depths of four feet. So long as the switch cap is secure the device may get wet and/or be submerged in water.
Does the device need to remain on the frame once the lengthening is complete?
Once the lengthening portion of the procedure has been completed, Automator 2000 may be removed from a ring frame and replaced by threaded rods. On a unilateral frame the second sliding component on the rail may be locked down and the device may be removed as well. Only the physician should remove the devices. The devices must be returned to Autogenesis, Inc. If you need furthur assistance, feel free to contact us by calling: